Removing the CO2 Scrubber 130 Removing the Pump 131. 7 ... M3015A and M3016A) 142 Exchange Parts List 145 Plug-in Modules Part Numbers 147 Part Number Table 147 Exchange Modules, Table 1 147 ... 9 Anesthetic Gas Module 187 10 Philips 15210B Calibration Unit 189 Description 189 Unpacking the Instrument 189
Apr 02, 2019· First time with the Philips monitor and we bought the MP20 with MMS and M3015a module for EtCO2. Everything is working fine except the awRR, it is ON but I cannot see anything on the screen. The CO2 measurement is OK and the number indication of the CO2 is OK. It says that I need to do CO2 calibration but it works fine without it.
hilips M3015A C02 MMS Module. The M3015A Microstream CO2 Extension is designed for use by clinicians for adult, pediatric and neonatal patients in Critical and Intermediate Care areas, in Operating Rooms and General Wards, and for transport inside and outside the hospital.
Philips Module M3015A Microstream CO2 Module for Patient Monitors. ABOUT-BPT MEDICAL LTD . We trusted medical supply company supplying new and used medical stock base in UK, we describe products and provide best after care including 30 days return policy to maintain high customer satisfaction and feedback standards.
Philips M3015A CO2 MMS Module. Overview; Videos; Description; Compatibles; View Options. Patient Monitoring; Monitors - Modules & Extensions; Philips M3015A CO2 MMS Module. Brand: Philips. Web ID: 14447. Condition: Refurbished. Product Number: UMPH7000. Select A Model (2 available) Philips M3015A Option C00 - C02 Only ...
Jan 15, 2020· The Philips module M3015a is an extension of the Philips multi-measurement server. This, in turn, measures capnography and has an additional port for monitoring the patient’s temperature or for invasive blood pressure measurement. This module has a number of benefits, including the ability to be added to the multi-measurement server on the go.
Philips Intellivue CO2 Module Sensor M3015a / M3015b P/N: 453564196491 This unit is cosmetically in very good condition, tubing still intact and connectors look good. This was pulled from a device that we used as a parts unit. We don’t currently have any way to test it so we are selling as is / untested.
The Philips IntelliVue M3015A Microstream CO2 Extension can be connected to Philips MMS Modules or IntelliVue X2. It provides low sample flow rate of CO2 measurements making it suitable for adult, pediatric, and neonatal patients in Critical and Intensive Care Units, Operating Rooms, Wards, and for general transport within and outside hospitals.
First time with the Philips monitor and we bought the MP20 with MMS and M3015a module for EtCO2. Everything is working fine except the awRR, it is ON but I cannot see anything on the screen. The CO2 measurement is OK and the number indication of the CO2 is OK. It says that I need to do CO2 calibration but it works fine without it.
Jun 08, 2020· Product. Philips M3015A Microstream CO2 Extension Model No. 862393 -used with an lntelliVue X3 Patient Monitor (Model No. 867030) or a lntelliVue MMX Multi-Measurement Module (Model No. 867036) Code Information. Units with this hardware version can be identified by the following serial number prefix: SN prefix: DE435xxxxx (first five digits)
Microstream CO2 is an innovative, advanced technology for sidestream capnography. The Philips Microstream CO2 Extension is designed to work with the Philips Multi-Measurement Module (MMS) and IntelliVue X2 and provides a real-time waveform and numeric values for CO2 on the Philips patient monitor display. Furthermore, Integrated Pulmonary Index™ (IPI)* helps caregivers to manage four ...
Oct 16, 2014· The Calibration Procedure for Philips M3015A EtCO2 Module Step 2 – Calibration Enter into service mode on the patient monitor and select Setup CO2. Connect the CO2 (5%) gas with the calibration line to the microstream CO2 port. This will activate the pump. Select Cal. CO2 and
Philips M3015A/B CO2 Module $ 2, Read more. Add to wishlist. Compare. Patient Monitoring. GE Carescape T14 Case Assembly. Read more. Add to wishlist. Compare. Patient Monitoring. GE Apex Pro Case Assembly. Read more. Add to wishlist. Compare. Patient Monitoring. Philips MX40 Telemetry Touch Screen $ Read more.
Mainstream CO2 measurements taken at the patients airway, resulting in faster response time and less chance of erroneous artefact data. Easy mounting of Sidestream CO2 sensor with integrated clip. A choice of disposable Sidestream CO2 accessories for both intubated and non-intubated applications.